There are two ways two generate Radiance calibration files for Field Spec Pro. One method uses a Panel, and the other method uses an Integrating Sphere.In the ‘Panel’ method, the illumination from a NIST traceable ‘Irradiance Source’ falls onto a nearly 100% reflectance, diffusepanel (White Reference Panel). Per NIST the panel is located 50cm away from the source. This is far enough away that the Flux density, or Irradiances at points a and c differ insignificantly from the Irradiance at point b. This also means that the distances from the Source to points a and c are insignificantly different from the distance, D from the Source to point b. The NIST traceable certificate provides the Source Irradiance values at point b for distance, D. The spectrometer input views an area safely within a and c. The source enclosure contains black baffles and apertures to prevent light from bouncing off surfaces and the calibration room (such bouncing light could contribute sources of error).
Ainda não possui cadastro?
Sign up for free today. By registering you will have free access to exclusive content including
- Webinars, presentations and videos
- Application notes, technical notes, articles, white papers and software downloads
And in addition you will receive
- Our regular eNews including the latest news, education, events and offers from Malvern Panalytical